1)No direct sunlight when using the instruments;
2)No poisonous ,flammable and corrosive gases when using the instruments;
3)Put the instruments on solid and flat platform ,avoiding shocks;
4)Pay attention to the local voltage fluctuation. If it is exceed the safety range, we suggest to add voltage-stabilized equipment;
5)Anyone without our company’s authorization is forbidden to disassembling ,maintain ,debug or replace any components or remove any wires. Otherwise we take no responsibility for the consequences.
6)Do not add on outside keyboard without permission, avoiding any damage on internal information or procedures .
7)Wipe the LCD screen with specialized soft cloth in your need.
Warning: Make the ECG cable away from neonate’s throat to avoid apnea.
Because neonate’s body is short and their skin is tender, please choose specialized neonatal ECG cable and electrode. And the electrode should be inspected and replaced periodly.
Warning: you may get the incorrect SPO2 value because of any movement during signal analysis, So make sure the patient remain as quiet as possible.
Choose the specialized neonatal SPO2 probe, do not cramp the probe on any damaged skin ,edema or fragile fingers, do not measuring over two hours on one single finger.
1)The fuse is broken.------------ change a fuse can solve the problem.
2)The inside wire of screen is loosen or fell off, the monitor with on display.
3)The system board failure, need to upgrade the software or change another system board.
4)The power module failure ,need to re-check or change another power module.
Possible causes to no NIBP value:
1).The patient over-exercised. ----keep patient calm down for a while.
2)The NIBP cuff is broken with air leakage. ---------change another cuff.
3.The NIBP module failure.--------need to re-check the module or replace a new one.
Possible causes to obviously incorrect NIBP value:
1) The patient over-exercised. ----keep patient calm down for a while.
2)The NIBP module failure.--------need to re-check the module or replace a new one.
Possible causes: The built-in Li-ion rechargeable battery is completely discharged or broken.
At this situation, leave the monitor open and get battery charged for at least one hour after adjust the system time correctly. Then close the monitor and restart again, if the system time still can’t be saved, change a new battery.